Open the Panel by selecting Panel in the View Selector. Expand the Build Array Node to add an extra terminal by dragging down with the mouse after seeing the expansion cursor appear on the bottom of the Node. Message Edited by kel on If you select the device, you can see additional properties, such as the device name, as well as any software or drivers that are installed on your system to support this device. Then click the green Run arrow at the top left of the VI. Use the following procedure to add manual inputs for Voltage, Current, and Resistance ranges Create an “Enable Auto Range” boolean control by right clicking the “Enable Auto Range” boolean constant and selecting Change to control.
|Date Added:||26 January 2015|
|File Size:||38.27 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Keithley Sub-Femtoamp Remote SourceMete – Instrument Driver – National Instruments
NI system SMUs combine power, precision, and speed into a single instrument. Drop down all of the indicators and controls from the Unplaced Items box onto the Panel. Let me know the results, and we’ll get to communicating! Message 4 of On the Build Array Node connect the first terminal Element 1 to the voltage indicator wire, Element 2 to the current indicator wire, and Element 3 to the resistance indicator wire.
Wire together the appended path terminal on the top right of the Build Path Node to the path terminal on the top left of the Write Delimited Spreadsheet Node. I got my drivers form the url: Select the number of desired bins in the histogram using the number of bins control.
Automating Measurements From a Keithley 2400 SourceMeter SMU Instrument With LabVIEW NXG
Customize the Panel of the VI to fit the application’s needs. Right click terminals and choose Create control to create a control wired to that terminal. Message 5 of Rename this indicator “New File Path”. Right Click the Autozero terminal and select Create control. If it’s only a few, you can just use VISA functions to directly command the instrument to do what you’d like!
First ensure you have the NI-Serial driver installed.
Does this return a description of the instrument? Usually a question mark being sent implies that the instrument will reply, which means you’ll want to read after you write the command. Labveiw that the above “major categories” are generally in the order in which they are called in a given program.
The COM port will not show up in the Hardware tab. Message 6 of Is there a way to program with strings that will achieve what I want I already do this with my Kepco power supply? What I need is to be able to set the Kepco power supply to 0 when the button is pressed aborting obviously won’t do this?
Then create an indicator for the new file terminal on the top right of the Write Delimiter Spreadsheet Node.
Use the following procedure to add manual inputs for Voltage, Current, and Resistance ranges Create an “Enable Auto Range” boolean control by right clicking the “Enable Auto Range” boolean constant and selecting Change to control.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
What functions are you interested in the instrument performing? Create the histogram graph by right clicking the histogram terminal on the top right hand side kabview the Histogram Node and selecting Create indicator. Applications Engineer National Instruments.
Does anybody have an example of a timed scan.
I seem to be having problems wiring a stop button to stop the program from running. If it’s only a few, you can just use VISA functions to directly command the instrument to do what you’d like! Thanks and looking forward to your repost!